SCORIA Wireless Air Quality Probe
SCORIA is indoor battery powered air quality measurement device, which uploads data to backend through various radio technologies such as NB-IoT or LoRaWAN.
|Device model name||SCORIA (previously: LX17TTS-SHT3X)|
Weight: ~200 grams
Dimensions: 125 x 100 x 35 mm
Enclosure material: white ABS plastic
|Power||Integrated rechargeable battery (charger order code: SCH-A0 for Li-Ion or SCH-A1 for LiFePO4 version)
Non-rechargeable battery, 1x C size 3.6V lithium battery
Recommended type: SAFT LS 26500 with 2x1 0,1” connector leads
Battery order code: ML-BATT-36-Mk1
|Typical battery life at ~23°C||Rechargeable battery (type: A0):
NB-IoT ~10.000 messages
~42.000 Sigfox messages
Wall adapter - no limit
|Standard operation environment||
-10 to +50 °C of ambient temperature
max. 90% of non-condeing relative humidity
|Wireless uplink options||NBIoT
Sigfox,Fully certified by Sigfox as “Sigfox Ready” device
Class 0u - the highest TX signal propagation class
Max radio power output: 14dBm / 25mW
Transmit cycle power usage ~320uWh @25mW TX power for cycle duration 15 minutes
|Antenna type||NB-IoT: Integrated antenna
LORAWAN: Integrated antenna
Sigfox: Custom in-house designed ISM trace antenna tuned for Sigfox network
3%RH for humidity
0.3 °C for temperature
±30ppm for CO2
Information included on the label:
- Device type
- Full order code
- Serial number
- Uplink identificators
Sigfox version information on label:
- Sigfox ID
- Sigfox PAC
LoRaWAN version information on label:
How to order
Order shall contain the following information:
- Date and internal number of the order
- Name and address (including VAT number where appropriate)
- Number of units and requested delivery date
- Type of delivery
Order code table
|Uplink radio type (R)|
|R1||LoRaWAN 1.0.2 - Class A/C - EU 868 MHz|
|R2||LoRaWAN 1.0.2 - Class A/C - US 915 MHz|
|R3||NB-IoT - band 20|
|R2||LoRaWAN AU 915-928 MHz|
|Type of measurement (T)|
|T1||Temperature, Humidity, CO2|
|T2||Temperature, Humidity, Barometric pressure|
|T3||Temperature, Humidity, CO2, Barometric pressure|
|T4||Temperature, Humidity, CO2, VOC|
|T6||Temperature, Humidity on 2m long cable|
|Power supply type (C)|
|C0||Non-rechargeable LiSOCl2 3.6V battery "C"|
|C1||Rechargeable Li-Ion battery (internal)|
|C2||Wall adapter (no battery)|
|C3||Rechargeable LiFePO4 battery (internal)|
Order code example: SCORIA-R3-T1-C0 Uplink radio type (R3): NBIoT - band 20 Type of measurement (T1): Temperature, Humidity, CO2 Power supply type (C0): Non-rechargeable LiSOCl2 3.6V battery "C"
Devices are packed individually into padded paper boxes.
The packaging includes all information required for proper use and servicing of the equipment.
The device must be installed on a vertical plane (wall) for optimal working conditions. Holes for intake of air pointing down and vent holes pointing up. The TOP of the device is either marked TOP or with an arrow pointing to the TOP of the device. TOP of the device should point upwards (towards ceiling).
For monitoring CO2 levels in classrooms or offices we recommend to install the sensor on the wall, minimum 1 meter distance from sitting person at the same hight as heads of people in the classroom or office.
Do not install facing direct sunlight.
Do not install into environments where relative air humidity stays above 80% RH for prolonged periods of time.
Device radio information is written on the packaging of the device. Serial ID is written on the device itself on the right edge of the device. Please use this serial number to communicate support requests with your supplier.
CO2 probe calibration
To adjust SCORIA CO2 probe against reference CO2 reading please execute following steps:
- Download "nRF Toolbox for Bluetooth LE" from Apple store or Google Play Store
- Connect to SCORIA device according to the serial number written on the sensor enclosure
- Tap "NUS/UART" and the console will launch
- Execute the following command (without quotes):
- "run frc 450" - where 450 is the reference value on the reference sensor
- The calibration procedure will take up to 3 minutes. Logging information will be displayed in the console.
- The sensor will reboot automatically after the calibration is finished.
EU DECLARATION OF CONFORMITY
Device type: SCORIA
Name and address of the manufacturer: Moire Labs s.r.o., Kulturna 32, 821 04 Bratislava, Slovakia
This declaration of conformity is issued under the sole responsibility of the manufacturer.
Object of declaration: Air Quality Sensor
The object of the declaration described herein is in conformity with the following Union harmonisation legislation:
- Directive 2014/53/EU (RED)
On behalf of Moire Labs in Bratislava, SK on 1st of December 2022: Marcel Hecko, CTO Moire Labs s.r.o.
Design and manufacture
Moire Labs s.r.o. Kultúrna 32 821 04 Bratislava Slovakia European Union
For support please contact your distributor or manufacturer directly via www.moirelabs.com